BREAKING NEWS
Phase noise

Summary

In signal processing, phase noise is the frequency-domain representation of random fluctuations in the phase of a waveform, corresponding to time-domain deviations from perfect periodicity (jitter). Generally speaking, radio-frequency engineers speak of the phase noise of an oscillator, whereas digital-system engineers work with the jitter of a clock.

Phase noise measured by signal source analyzer (SSA). The SSA shows the positive part of the phase noise. In this picture there is a phase noise of the main carrier, 3 other signals and "noise hill".
A weak signal disappears in the phase noise of the stronger signal

Definitions

Historically there have been two conflicting yet widely used definitions for phase noise. Some authors define phase noise to be the spectral density of a signal's phase only,[1] while the other definition refers to the phase spectrum (which pairs up with the amplitude spectrum) resulting from the spectral estimation of the signal itself.[2] Both definitions yield the same result at offset frequencies well removed from the carrier. At close-in offsets however, the two definitions differ.[3]

The IEEE defines phase noise as ℒ(f) = Sφ(f)/2 where the "phase instability" Sφ(f) is the one-sided spectral density of a signal's phase deviation.[4] Although Sφ(f) is a one-sided function, it represents "the double-sideband spectral density of phase fluctuation".[5][clarification needed] The symbol is called a (capital or uppercase) script L.[6]

Background

An ideal oscillator would generate a pure sine wave. In the frequency domain, this would be represented as a single pair of Dirac delta functions (positive and negative conjugates) at the oscillator's frequency; i.e., all the signal's power is at a single frequency. All real oscillators have phase modulated noise components. The phase noise components spread the power of a signal to adjacent frequencies, resulting in noise sidebands. Oscillator phase noise often includes low frequency flicker noise and may include white noise.

Consider the following noise-free signal:

v(t) = Acos(2πf0t).

Phase noise is added to this signal by adding a stochastic process represented by φ to the signal as follows:

v(t) = Acos(2πf0t + φ(t)).

Phase noise is a type of cyclostationary noise and is closely related to jitter, a particularly important type of phase noise is that produced by oscillators.

Phase noise (ℒ(f)) is typically expressed in units of dBc/Hz, and it represents the noise power relative to the carrier contained in a 1 Hz bandwidth centered at a certain offsets from the carrier. For example, a certain signal may have a phase noise of −80 dBc/Hz at an offset of 10 kHz and −95 dBc/Hz at an offset of 100 kHz. Phase noise can be measured and expressed as single-sideband or double-sideband values, but as noted earlier, the IEEE has adopted the definition as one-half of the double-sideband PSD.

Jitter conversions

Phase noise is sometimes also measured and expressed as a power obtained by integrating ℒ(f) over a certain range of offset frequencies. For example, the phase noise may be −40 dBc integrated over the range of 1 kHz to 100 kHz. This integrated phase noise (expressed in degrees) can be converted to jitter (expressed in seconds) using the following formula:

${\displaystyle {\text{jitter (seconds}})={\frac {{\text{phase error (}}{}^{\circ }{\text{)}}}{360^{\circ }\times {\text{frequency (hertz)}}}}}$

In the absence of 1/f noise in a region where the phase noise displays a –20 dBc/decade slope (Leeson's equation), the RMS cycle jitter can be related to the phase noise by:[7]

${\displaystyle \sigma _{c}^{2}={\frac {f^{2}{\mathcal {L}}\left(f\right)}{f_{\text{osc}}^{3}}}}$

Likewise:

${\displaystyle {\mathcal {L}}\left(f\right)={\frac {f_{\text{osc}}^{3}\sigma _{c}^{2}}{f^{2}}}}$

Measurement

Phase noise can be measured using a spectrum analyzer if the phase noise of the device under test (DUT) is large with respect to the spectrum analyzer's local oscillator. Care should be taken that observed values are due to the measured signal and not the shape factor of the spectrum analyzer's filters. Spectrum analyzer based measurement can show the phase-noise power over many decades of frequency; e.g., 1 Hz to 10 MHz. The slope with offset frequency in various offset frequency regions can provide clues as to the source of the noise; e.g., low frequency flicker noise decreasing at 30 dB per decade (= 9 dB per octave).[8]

Phase noise measurement systems are alternatives to spectrum analyzers. These systems may use internal and external references and allow measurement of both residual (additive) and absolute noise. Additionally, these systems can make low-noise, close-to-the-carrier, measurements.

Spectral purity

The sinewave output of an ideal oscillator is a single line in the frequency spectrum. Such perfect spectral purity is not achievable in a practical oscillator. Spreading of the spectrum line caused by phase noise must be minimised in the local oscillator for a superheterodyne receiver because it defeats the aim of restricting the receiver frequency range by filters in the IF (intermediate frequency) amplifier.

References

1. ^ Rutman, J.; Walls, F. L. (June 1991), "Characterization of frequency stability in precision frequency sources" (PDF), Proceedings of the IEEE, 79 (6): 952–960, Bibcode:1991IEEEP..79..952R, doi:10.1109/5.84972, archived (PDF) from the original on 2022-10-09
2. ^ Demir, A.; Mehrotra, A.; Roychowdhury, J. (May 2000), "Phase noise in oscillators: a unifying theory and numerical methods for characterization" (PDF), IEEE Transactions on Circuits and Systems I: Fundamental Theory and Applications, 47 (5): 655–674, CiteSeerX 10.1.1.335.5342, doi:10.1109/81.847872, ISSN 1057-7122, archived (PDF) from the original on 2022-10-09
3. ^ Navid, R.; Jungemann, C.; Lee, T. H.; Dutton, R. W. (2004), "Close-in phase noise in electrical oscillators", Proc. SPIE Symp. Fluctuations and Noise, Maspalomas, Spain
4. ^ Vig, John R.; Ferre-Pikal, Eva. S.; Camparo, J. C.; Cutler, L. S.; Maleki, L.; Riley, W. J.; Stein, S. R.; Thomas, C.; Walls, F. L.; White, J. D. (26 March 1999), IEEE Standard Definitions of Physical Quantities for Fundamental Frequency and Time Metrology – Random Instabilities, IEEE, ISBN 978-0-7381-1754-6, IEEE Std 1139-1999, see definition 2.7.
5. ^ IEEE 1999, p. 2, stating ℒ(f) "is one half of the double-sideband spectral density of phase fluctuations."
6. ^ IEEE 1999, p. 2
7. ^ An Overview of Phase Noise and Jitter (PDF), Keysight Technologies, May 17, 2001, archived (PDF) from the original on 2022-10-09
8. ^ Cerda, Ramon M. (July 2006), "Impact of ultralow phase noise oscillators on system performance" (PDF), RF Design: 28–34, archived (PDF) from the original on 2022-10-09

• Rubiola, Enrico (2008), Phase Noise and Frequency Stability in Oscillators, Cambridge University Press, ISBN 978-0-521-88677-2
• Wolaver, Dan H. (1991), Phase-Locked Loop Circuit Design, Prentice Hall, ISBN 978-0-13-662743-2
• Lax, M. (August 1967), "Classical noise. V. Noise in self-sustained oscillators", Physical Review, 160 (2): 290–307, Bibcode:1967PhRv..160..290L, doi:10.1103/PhysRev.160.290
• Hajimiri, A.; Lee, T. H. (February 1998), "A general theory of phase noise in electrical oscillators" (PDF), IEEE Journal of Solid-State Circuits, 33 (2): 179–194, Bibcode:1998IJSSC..33..179H, doi:10.1109/4.658619, archived from the original (PDF) on 2015-03-05, retrieved 2021-09-16
• Pulikkoonattu, R. (June 12, 2007), Oscillator Phase Noise and Sampling Clock Jitter (PDF), Tech Note, Bangalore, India: ST Microelectronics, archived (PDF) from the original on 2022-10-09, retrieved March 29, 2012
• Chorti, A.; Brookes, M. (September 2006), "A spectral model for RF oscillators with power-law phase noise" (PDF), IEEE Transactions on Circuits and Systems I: Regular Papers, 53 (9): 1989–1999, doi:10.1109/TCSI.2006.881182, hdl:10044/1/676, S2CID 8855005, archived (PDF) from the original on 2022-10-09
• Rohde, Ulrich L.; Poddar, Ajay K.; Böck, Georg (May 2005), The Design of Modern Microwave Oscillators for Wireless Applications, New York, NY: John Wiley & Sons, ISBN 978-0-471-72342-4
• Ulrich L. Rohde, A New and Efficient Method of Designing Low Noise Microwave Oscillators, https://depositonce.tu-berlin.de/bitstream/11303/1306/1/Dokument_16.pdf
• Ajay Poddar, Ulrich Rohde, Anisha Apte, “ How Low Can They Go, Oscillator Phase noise model, Theoretical, Experimental Validation, and Phase Noise Measurements”, IEEE Microwave Magazine, Vol. 14, No. 6, pp. 50–72, September/October 2013.
• Ulrich Rohde, Ajay Poddar, Anisha Apte, “Getting Its Measure”, IEEE Microwave Magazine, Vol. 14, No. 6, pp. 73–86, September/October 2013
• U. L. Rohde, A. K. Poddar, Anisha Apte, “Phase noise measurement and its limitations”, Microwave Journal, pp. 22–46, May 2013
• A. K. Poddar, U.L. Rohde, “Technique to Minimize Phase Noise of Crystal Oscillators”, Microwave Journal, pp. 132–150, May 2013.
• A. K. Poddar, U. L. Rohde, and E. Rubiola, “Phase noise measurement: Challenges and uncertainty”, 2014 IEEE IMaRC, Bangalore, Dec 2014.